normalizePlates {cellHTS2} | R Documentation

## Per-plate data transformation, normalization and variance adjustment

### Description

Plate-by-plate normalization of the raw data stored in slot `assayData` of a `cellHTS` object. Normalization is performed separately for each plate, replicate and channel. `Log2` data transformation can be performed and variance adjustment can be performed in different ways (none, per-plate, per-batch or per-experiment).

### Usage

    normalizePlates(object,  scale="additive", log=FALSE, method="median", varianceAdjust="none", posControls, negControls,...)

### Arguments

`object` | a `cellHTS` object that has already been configured. See details.
`scale` | a character specifying the scale that the input data are considered to be on: "additive" scale (default) or "multiplicative". The interpretation of this terminology is that data on an additive scale will be normalised by subtraction of a correction offset, whereas data on a multiplicative scale are normalised by division through a correction factor.
`log` | logical. If `TRUE`, data will first be `log2` transformed. If data are on an additive scale (i.e. if `scale` is `"additive"`), then `log` is only allowed to be `FALSE`. The default is `log=FALSE`.
`method` | character specifying the normalization method to use for the per-plate normalization. Allowed values are `"median"` (the default), `"mean"`, `"shorth"`, `"POC"`, `"NPI"`, `"negatives"`, `Bscore` and `"locfit"`. See details.
`varianceAdjust` | character specifying the variance adjustment to perform. Allowed values are `"none"` (the default), code"byPlate", `"byBatch"` and `"byExperiment"`. See details.
`posControls` | a vector of regular expressions giving the name of the positive control(s). See details.
`negControls` | a vector of regular expressions giving the name of the negative control(s). See details.
`...` | Further arguments that get passed on to the function implementing the normalization method chosen by `method`. Currently, this is only used for `Bscore` and `locfit`.

### Details

The function `normalizePlates` uses the content of the `assayData` slot of `object`. For dual-channel data, a recommended workflow is (i) to correct for plate effects using the `normalizePlates` function, (ii) combine the two channels using the function `summarizeChannels`, and (iii) finally, if necessary, normalize the summarized intensities calling `normalizePlates` again.

In this function, the normalization is performed in a plate-by-plate fashion, following this workflow:

1. Log transformation of the data (optional)

2. Per-plate normalization

3. Variance adjustment of the plate intensity corrected data (optional)

The argument `scale` defines the scale of the data. If the data are on a multiplicative scale (`scale="multiplicative"`), the data can be `log2` transformed by setting `log=TRUE`. This then changes the scale of the data to code"additive".

In the next step of preprocessing, intensities are corrected in a plate-by-plate basis using the chosen normalization method:

- If `method="median"`, plates effects are corrected by the median value across wells that are annotated as `sample` in `wellAnno(object)`, for each plate and replicate.

- If `method="mean"`, the average in the `sample` wells is used instead.

- If `method="shorth"`, the midpoint of the `shorth` of the distribution of values in the wells annotated as `sample` is used.

- If `method="negatives"`, the median of the negative controls is used.

Depending on the scale of the data prior to normalization, the data are divided by the above defined correction factors (scale: `"multiplicative"`), or the value is subtracted (scale: `"additive"`).

Further available normalization methods are:

- `method="POC"` (percent of control): for each plate and replicate, each measurement is divided by the average of the measurements on the plate positive controls, and multiplied by 100.

- `method="NPI"` (normalized percent inhibition): each measurement is subtracted from the average of the intensities on the plate positive controls, and this result is divided by the difference between the means of the measurements on the positive and the negative controls.

- `method="Bscore"`: for each plate and replicate, the `B-score method`, which is based on a 2-way median polish, is applied to remove row and column biases.

- `method="locfit"` (robust local fit regression): for each plate and replicate, spatial effects are removed by fitting a bivariate local polynomial regression (see `spatialNormalization`).

In the final preprocessing step, variance of plate-corrected intensities can be adjusted as follows:

- `varianceAdjust="byPlate"`: per plate normalized intensities are divided by the per-plate median absolute deviations (MAD) in "sample" wells. This is done separately for each replicate and channel;

- `varianceAdjust="byBatch"`: using the content of slot `batch`, plates are split according to assay batches and the individual normalized intensities in each group of plates (batch) are divided by the per-batch of plates MAD values (calculated based on "sample" wells). This is done separately for each replicate and channel;

- `varianceAdjust="byExperiment"`: each normalized measurement is divided by the overall MAD of normalized values in wells containing "sample". This is done separately for each replicate and channel;

By default, no variance adjustment is performed (`varianceAdjust="none"`).

The arguments `posControls` and `negControls` are required for applying the normalization methods based on the control measurements that is, when `method="POC"`, or `method="NPI"`, or `method="negatives"`). `posControls` and `negControls` should be vectors of regular expression patterns specifying the name of the positive(s) and negative(s) controls, respectivey, as provided in the plate configuration file (and accessed via `wellAnno(object)`). The length of these vectors should be equal to the current number of channels in `object` (i.e. to the `dim(Data(object))[3]`). By default, if `posControls` is not given, pos will be taken as the name for the wells containing positive controls. Similarly, if `negControls` is missing, by default neg will be considered as the name used to annotate the negative controls. The content of `posControls` and `negControls` will be passed to `regexpr` for pattern matching within the well annotation given in the featureData slot of `object` (which can be accessed via `wellAnno(object)`) (see examples for `summarizeChannels`). The arguments `posControls` and `negControls` are particularly useful in multi-channel data since the controls might be reporter-specific, or after normalizing multi-channel data.

See the Examples section for an example on how this function can be used to apply a robust version of the Z score method, whereby, for each plate and replicate, the per-plate median (computed only from sample wells) is subtracted from the measurements, and the result is divided by the per-plate MAD (only from sample wells).

### Value

An object of class `cellHTS` with the normalized data stored in slot `assayData` (its previous contents were overridden). The processing status of the `object` is updated in the slot `state` to `object@state[["normalized"]]=TRUE`.

Additional slots of `object` may be updated if `method="Bscore"` or `method="locfit"` are used. Please refer to the help page of the `Bscore` function and `spatialNormalization` functions.

### Author(s)

Ligia Bras ligia@ebi.ac.uk, Wolfgang Huber huber@ebi.ac.uk

### References

Boutros, M., Bras, L.P. and Huber, W. (2006) Analysis of cell-based RNAi screens, Genome Biology 7, R66.

### See Also

`Bscore`, `spatialNormalization`, `summarizeChannels`

### Examples

        data(KcViabSmall)
        # per-plate median scaling of intensities
        x1 <- normalizePlates(KcViabSmall, scale="multiplicative", log=FALSE, method="median", varianceAdjust="none")
        # per-plate median subtraction of log2 transformed intensities
        x2 <- normalizePlates(KcViabSmall, scale="multiplicative", log=TRUE, method="median", varianceAdjust="none")
        ## Not run:
        x3 <- normalizePlates(KcViabSmall, scale="multiplicative", log=TRUE, method="Bscore", varianceAdjust="none", save.model=TRUE)

    ## End(Not run)

        ## robust Z score method (plate intensities are subtracted by the per-plate median on sample wells and divided by the per-plate MAD on sample wells):
        xZ <- normalizePlates(KcViabSmall, scale="additive", log=FALSE, method="median", varianceAdjust="byPlate")

        ## an example to illustrate the use of slot 'batch':
       ## Not run:
       try(xnorm <- normalizePlates(KcViabSmall, scale="multiplicative", method="median", varianceAdjust="byBatch"))

       # It doesn't work because we need to have slot 'batch'!
       # For example, we will suppose that a different lot of reagents was used for plate 1:
       pp <- plate(KcViabSmall)
       fData(KcViabSmall)$"reagent" <- "lot B"
       fData(KcViabSmall)$"reagent"[pp==1] <- "lot A"
       fvarMetadata(KcViabSmall)["reagent",] <- "Lot of reagent used"

       bb <- as.factor(fData(KcViabSmall)$"reagent")
       batch(KcViabSmall) <- array(as.integer(bb), dim=dim(Data(KcViabSmall)))
       ## check number of batches:
       nbatch(KcViabSmall)
       x1 <- normalizePlates(KcViabSmall, scale="multiplicative", log = FALSE, method="median", varianceAdjust="byBatch")

    ## End(Not run)

[Package cellHTS2 version 2.52.0 Index]